|Product Name:||Quick Release Drop Tester||Power Required:||AC220V / 50Hz|
|Compressed Air:||0.5 Mpa-0.7 Mpa||Lifting Device:||User Provided|
|Machine Weight:||16 Kg||Machine Size:||345 X 235 X 145 Mm|
quick release drop tester,
quick release hook drop test
1000KG Quick Release Drop Tester Designed To Drop Large Packaging Meet ISTA Standard
|Power Required||AC220V / 50Hz ( or according to customer’s need)|
|Lifting Device||Customer Supplied|
|Weight of KDT2000||11kg ( excluding foot-switch, which is 5kg)|
|Size of KDT2000||345 x 235 x 145 mm|
|Size of foot-switch||
270 x 200 x220 mm ( Lx D x H)
Release hooks help determine the ability of a product or container to withstand impact loads which may be imposed upon such items during normal transit conditions. Labtone Release Hooks can occommodate large and bulky items. Our full line of Release Hooks comply with ISTA, ASTM, ISO, and other internationally recognized standards.
The KDT2000 Drop Tester is designed to simulate the drops and mechanical hazards that occur in distribution. It is especially suited to very large, heavy payloads which cannot be tested on conventional drop test machines.Regardless of the size or shape of the package, the system is capable of performing a variety of free fall(flat, corner or edge). When used, lift the ring by lifting it onto the release mechanism jaw and then lifting the release mechanism and the test package up to the test drop height by lifting the device. The foot release switch activates the solenoid valve in the release mechanism so that the release mechanism jaws open and the package is free to fall.
1) Technical consultation: test method, laboratory planning and suggestion.
2) Equipment selection: selection scheme, FAQ.
3) Product testing scheme.
4) Customer communication and progress report.
1) Technical training: operation of equipment, daily maintenance, common fault diagnosis and troubleshooting.
2) Scheduled on-site service: Detection of problem as soon as possible in order to eliminate equipment and anthropogenic hazards. This is to ensure long term and stable equipment operation as well as delivery of latest technical information.
3) Technical support: special paid services are provided according to customer’s needs.
How do you attach the hardware you’re testing to your shaker?
By means of a fixture, usually aluminum or magnesium for lightness coupled with rigidity. They can be cast, or smaller fixtures machined from solid stock. Most fixtures are welded.
How do you control shakers?
If we’re looking for resonances in the product we’re testing, we command the shaker to shake the product at one frequency at a time but to vary that test frequency, to sweep it over a range of frequencies.But more realistically, we command the shaker to vibrate randomly and to excite all the resonances simultaneously.Control commands go into the keyboard of a specially-programmed computer.
What are those resonances?Are they bad?
Have you ever noticed the steering wheel moving with rather large displacement amplitude, larger than the input to the column? That magnification is called resonance. Possibly it annoys you. There’s a slight chance that in a few years that whipping of the steering column might cause bending fatigue failure.When we shake an automotive or ship or land vehicle instrument, we're looking for, for example, portions of printed wiring boards (PWBs) responding with greater motion than we're inputting. That flexing may damage PWB wiring, it may damage the attached components, and it will damage the soldered connections between components and the PWB.
Packaging & Shipping
Simulation Animatronic Customer Triceratops are covered with air bubble film before put them into the wooden case,which not only has good shock absorption, impact resistance, heat sealing and also has the advantages of nontoxic, odorless,moisture corrosion, good transparency etc
Contact Person: Yolin Yang